DCM 320/330/340/1500A – Clamp Meters
AVO210 & 410 - Multi Meters
SAFETY ALERT – Low Battery Warning
Safety is always Megger’s highest priority and as such we wish to draw your attention to an issue that has been identified on the above products.
Once the low battery indicator appears, readings may become unreliable to a point where the instrument can potentially show 0 V when a voltage is present, typically when the battery supply voltage drops below 50%. Accordingly:
To avoid false readings that could lead to electric shock and injury, do not use the product for measurement when the low battery warning is on. Batteries must be replaced as soon as the low battery indicator appears to avoid unreliable or false readings.
Please ensure you are using the latest version of the manual for the above instruments.
If you have any questions regarding this matter please contact the Technical Support Team at Megger, Dover or your regional Technical Support Team, as detailed below.
Europe, Middle East & Africa
+44 (0)1304 502102
Asia & Pacific
+63 977 856 7182
Group Operations Director